دستگاه سختی سنج HARTIP 2500 دستگاه سختی سنج HARTIP 2500 یکی از جدیدترین دستگاه های سختی سنج پرتابل با دقت 2 HL or 03 HL800 میباشد   صفحه نمایش بزرگ و رنگی  دقت بالا و تکرار پذیری بالا از محسنات این دستگاه سختی سنج میباشد   Advantages   Patent dual coil technology  Equipped with digital probes only  Cost effec tive with reading probe optional  High repeatability accuracy 2 HL or 03 HL800  High linear accuracy for any angles  TFT color display  Auto compensate error for different impac t direc tion  Powered by AA batter y or USB power supply  The tester can be attrac ted to the workpiece with magnetic base optional  Statistics value can be calculated automatically  Up to 10 kinds of menu language  Data managing sof tware on PC  Power onoff automatically or manually  Specification Principle 	Leeb hardness measurement Accuracy  Repeatability 	03  HL800  2HL Display High contrast color 	TFT 240 x 320 Hardness scale 	HLHRCHRBHBHVHSHRAb Measuring range 	HL100960  HRC09792  HRB12140  HB11878   HV11698  HS05128  HRA7885  brm246599Nmm2 Probe 	Digital probe D CDCD15DLE  Wireless probe D CDCD15DLE optional Reading probe D optional Materials and conversion 	curves 11 common metal materials 75 curves Memory 999 data 	999 data Statistics function 	Average  Max  Min  S value Recalibration 	Allowed by user Indicator 	Low battery Language 	En g l i s h  S i m p l i f i e d Chinese Trad i tiona l Chinese Russian   Korean Fre nch  Italian  German Turkish  Spanish Interface USB  Bluetooth optional 	USB  Bluetooth optional Power onoff 	Auto  Manual Power supply 	AA alkaline battery  USB power supply Working environment 	20C70C Dimension mm 1 	145x70x30 Net weight g 	145 Standards 	Conforming to ASTM A956 DIN 50156 GBT 173941998